Mijn klantenzone
Inloggen
Registreren
Home
Downloads
Over ons
FAQ
Contact
Raamovereenkomst
Vacature
nl
fr
en
de
Winkel
mandje
0
U bent hier:
Home
Labo elektriciteit
Meetapparatuur Labo ELektr.
Watt-Volt-Var meters
Watt-Volt-Var meters
22/22 producten
Sorteer
Naam A-Z
Naam Z-A
Prijs laag-hoog
Prijs hoog-laag
Recentst
Populairst
Alle producten
Aantal per pagina: 50
Aantal per pagina: 100
Aantal per pagina: 200
10A-20A Varmeter (100 - 200 - 400 V)
VAR05/ELE
" VAR varmeters measure the reactive power absorbed by a 50Hz sinusoidal curve.
" protected against any overloads in their current
and voltage circuits, by fuses.
" Scale length: 110mm graduated from 0 to 100.
" Dimensions: 154 x 210 x 86mm.
" Weight: 1.5kg.
I (Imax=1,5 x I) : 10A - 20 A
V (Vmax=1,5 x V): 100 - 200 - 400 V
Prijs op aanvraag
Meer info ›
Toevoegen aan verlanglijst ›
5A-10A Varmeter (100 - 200 - 400V)
VAR04/ELE
" VAR varmeters measure the reactive power absorbed by a 50Hz sinusoidal curve.
" protected against any overloads in their current
and voltage circuits, by fuses.
" Scale length: 110mm graduated from 0 to 100.
" Dimensions: 154 x 210 x 86mm.
" Weight: 1.5kg.
I (Imax=1,5 x I) : 5A - 10 A
V (Vmax=1,5 x V): 100 - 200 - 400 V
Prijs op aanvraag
Meer info ›
Toevoegen aan verlanglijst ›
2.5A-5A Varmeter (100 - 200 - 400V)
VAR03/ELE
" VAR varmeters measure the reactive power absorbed by a 50Hz sinusoidal curve.
" protected against any overloads in their current
and voltage circuits, by fuses.
" Scale length: 110mm graduated from 0 to 100.
" Dimensions: 154 x 210 x 86mm.
" Weight: 1.5kg.
I (Imax=1,5 x I) : 2,5A - 5 A
V (Vmax=1,5 x V): 100 - 200 - 400 V
Prijs op aanvraag
Meer info ›
Toevoegen aan verlanglijst ›
1A-2A Varmeter (100 - 200 - 400V)
VAR02/ELE
" VAR varmeters measure the reactive power absorbed by a 50Hz sinusoidal curve.
" protected against any overloads in their current
and voltage circuits, by fuses.
" Scale length: 110mm graduated from 0 to 100.
" Dimensions: 154 x 210 x 86mm.
" Weight: 1.5kg.
I (Imax=1,5 x I) : 1A - 2 A
V (Vmax=1,5 x V): 100 - 200 - 400 V
Prijs op aanvraag
Meer info ›
Toevoegen aan verlanglijst ›
500mA - 1A Varmeter (100 - 200 - 400V)
VAR01/ELE
" VAR varmeters measure the reactive power absorbed by a 50Hz sinusoidal curve.
" protected against any overloads in their current
and voltage circuits, by fuses.
" Scale length: 110mm graduated from 0 to 100.
" Dimensions: 154 x 210 x 86mm.
" Weight: 1.5kg.
I (Imax=1,5 x I) : 500 mA - 1 A
V (Vmax=1,5 x V): 100 - 200 - 400 V
Prijs op aanvraag
Meer info ›
Toevoegen aan verlanglijst ›
Magnetoelectric Voltmeter - 30mV-600V (DC) ; 1.5%
PSY600V/ELE
This magneto-electric voltmeter measures DC and sinusoidal AC voltages. The apparatus has full overload protection by gas tube and an ultra-fast response time fuse with a low resistance to avoid changes in precision as a result of replacement.
EASY TO READ
To use the same scale for both AC and DC the PSY600V is equipped with a stepup transformer. The 0-100 scale is located above the antiparallax mirror, the 0-30 scale (with 60 divisions) below it.
OTHERS CHARACTERISTICS
" Safety terminals
" Anti-flash switch with silver contacts
" The 2mm space between contacts protects the switch from arcing due to voltage overload.
" Taut band mechanism
" Scale length 125 mm with antiparallax mirror
" Dimensions 196 x 147 x 70 mm.
" Weight 1.4kg
DC:
30-100-300mV - 1-3-10-30-100-300-600V
Accuracy 1.5% / Impedance 30k©/V
AC:
1-3-10-30-100-300-600V
Accuracy 1,5% / Impedance 5k©/V
PROTECTED yes
Prijs op aanvraag
Meer info ›
Toevoegen aan verlanglijst ›
RMS AC Wattmeter 500mA-1-2.5-5-10-25A (single phase, 3-phase)
PSY16/ELE
The PSY wattmeters measure true RMS and instantaneous power,
DC, single phase, balanced three-phase, alternating signal with
or without DC offset in the frequency range DC up to 2kHz.
ACCURACY : 0.5 %
PROTECTION
" All PSY wattmeters are guaranted within their accuracy limits
for 5 years.
" A fuse to protect the voltage circuit and another for the current.
" The fuses can be accessed from the outside of the meter.
EASY TO READ
" A single scale graduated from 0 to 100.
" The reading on the scale shows the maximum values:
I max = 1.5 x I V max = 1.5 x V
" A lookup table for three phase power can be found on the
back of the unit.
EXAMPLE READING
" For instance we wish to measure the power dissapated in a
load with 0.6A at 125V.
" Put the selector knobs to 1A and 100V.
" The needle moves to the marking corresponding to 75W.
" The reading is direct.
" The 100V scale is not overloaded.
(V max = 1.5 x V therefore 150V).
GENERAL SPECIFICATIONS
" Voltage circuit resistance: 150©/V
" Length of scale: 110 mm
" Dimensions 154 x 210 x 86mm.
" Weight: 1.5kg
" CEI1010 CATIII 1000Vrms pol2
PSY16
CONTINUOUS - SINGLE PHASE
I (I max = 1,5 x I): 500mA-1-2,5-5-10-25A
V (V max = 1,5 x V): 10 - 20 - 50 - 100 - 200 - 400V
3-PHASE
V (V max = 1,5 x V): 380 - 220 - 100 - 50 - 20V
INTENSITY FUSE: /
Prijs op aanvraag
Meer info ›
Toevoegen aan verlanglijst ›
RMS AC+DC Wattmeter 10 - 20A (continous, single phase, 3-phase)
PSY15/ELE
The PSY wattmeters measure true RMS and instantaneous power,
DC, single phase, balanced three-phase, alternating signal with
or without DC offset in the frequency range DC up to 2kHz.
ACCURACY : 0.5 %
PROTECTION
" All PSY wattmeters are guaranted within their accuracy limits
for 5 years.
" A fuse to protect the voltage circuit and another for the current.
" The fuses can be accessed from the outside of the meter.
EASY TO READ
" A single scale graduated from 0 to 100.
" The reading on the scale shows the maximum values:
I max = 1.5 x I V max = 1.5 x V
" A lookup table for three phase power can be found on the
back of the unit.
EXAMPLE READING
" For instance we wish to measure the power dissapated in a
load with 0.6A at 125V.
" Put the selector knobs to 1A and 100V.
" The needle moves to the marking corresponding to 75W.
" The reading is direct.
" The 100V scale is not overloaded.
(V max = 1.5 x V therefore 150V).
GENERAL SPECIFICATIONS
" Voltage circuit resistance: 150©/V
" Length of scale: 110 mm
" Dimensions 154 x 210 x 86mm.
" Weight: 1.5kg
" CEI1010 CATIII 1000Vrms pol2
PSY15
CONTINUOUS - SINGLE PHASE
I (I max = 1,5 x I): 10 - 20A
V (V max = 1,5 x V): 10 - 20 - 50 - 100 - 200 - 400V
3-PHASE
V (V max = 1,5 x V): 380 - 220 - 100 - 50 - 20V
INTENSITY FUSE: /
Prijs op aanvraag
Meer info ›
Toevoegen aan verlanglijst ›
WATTMETER RMS AC+DC 5 - 10 A
PSY14/ELE
The PSY wattmeters measure true RMS and instantaneous power,
DC, single phase, balanced three-phase, alternating signal with
or without DC offset in the frequency range DC up to 2kHz.
ACCURACY : 0.5 %
PROTECTION
" All PSY wattmeters are guaranted within their accuracy limits
for 5 years.
" A fuse to protect the voltage circuit and another for the current.
" The fuses can be accessed from the outside of the meter.
EASY TO READ
" A single scale graduated from 0 to 100.
" The reading on the scale shows the maximum values:
I max = 1.5 x I V max = 1.5 x V
" A lookup table for three phase power can be found on the
back of the unit.
EXAMPLE READING
" For instance we wish to measure the power dissapated in a
load with 0.6A at 125V.
" Put the selector knobs to 1A and 100V.
" The needle moves to the marking corresponding to 75W.
" The reading is direct.
" The 100V scale is not overloaded.
(V max = 1.5 x V therefore 150V).
GENERAL SPECIFICATIONS
" Voltage circuit resistance: 150©/V
" Length of scale: 110 mm
" Dimensions 154 x 210 x 86mm.
" Weight: 1.5kg
" CEI1010 CATIII 1000Vrms pol2
PSY14
CONTINUOUS - SINGLE PHASE
I (I max = 1,5 x I): 5 - 10A
V (V max = 1,5 x V): 10 - 20 - 50 - 100 - 200 - 400V
3-PHASE
V (V max = 1,5 x V): 380 - 220 - 100 - 50 - 20V
INTENSITY FUSE: 16A temp.
Prijs op aanvraag
Meer info ›
Toevoegen aan verlanglijst ›
RMS AC+DC Wattmeter 2.5 - 5A (continous, single phase, 3-phase)
PSY13/ELE
The PSY wattmeters measure true RMS and instantaneous power,
DC, single phase, balanced three-phase, alternating signal with
or without DC offset in the frequency range DC up to 2kHz.
ACCURACY : 0.5 %
PROTECTION
" All PSY wattmeters are guaranted within their accuracy limits
for 5 years.
" A fuse to protect the voltage circuit and another for the current.
" The fuses can be accessed from the outside of the meter.
EASY TO READ
" A single scale graduated from 0 to 100.
" The reading on the scale shows the maximum values:
I max = 1.5 x I V max = 1.5 x V
" A lookup table for three phase power can be found on the
back of the unit.
EXAMPLE READING
" For instance we wish to measure the power dissapated in a
load with 0.6A at 125V.
" Put the selector knobs to 1A and 100V.
" The needle moves to the marking corresponding to 75W.
" The reading is direct.
" The 100V scale is not overloaded.
(V max = 1.5 x V therefore 150V).
GENERAL SPECIFICATIONS
" Voltage circuit resistance: 150©/V
" Length of scale: 110 mm
" Dimensions 154 x 210 x 86mm.
" Weight: 1.5kg
" CEI1010 CATIII 1000Vrms pol2
PSY13
CONTINUOUS - SINGLE PHASE
I (I max = 1,5 x I): 2,5 - 5A
V (V max = 1,5 x V): 10 - 20 - 50 - 100 - 200 - 400V
3-PHASE
V (V max = 1,5 x V): 380 - 220 - 100 - 50 - 20V
INTENSITY FUSE: 8A temp.
Prijs op aanvraag
Meer info ›
Toevoegen aan verlanglijst ›
RMS AC+DC Wattmeter 1 - 2 A (continous, single phase, 3-phase)
PSY12/ELE
The PSY wattmeters measure true RMS and instantaneous power,
DC, single phase, balanced three-phase, alternating signal with
or without DC offset in the frequency range DC up to 2kHz.
ACCURACY : 0.5 %
PROTECTION
" All PSY wattmeters are guaranted within their accuracy limits
for 5 years.
" A fuse to protect the voltage circuit and another for the current.
" The fuses can be accessed from the outside of the meter.
EASY TO READ
" A single scale graduated from 0 to 100.
" The reading on the scale shows the maximum values:
I max = 1.5 x I V max = 1.5 x V
" A lookup table for three phase power can be found on the
back of the unit.
EXAMPLE READING
" For instance we wish to measure the power dissapated in a
load with 0.6A at 125V.
" Put the selector knobs to 1A and 100V.
" The needle moves to the marking corresponding to 75W.
" The reading is direct.
" The 100V scale is not overloaded.
(V max = 1.5 x V therefore 150V).
GENERAL SPECIFICATIONS
" Voltage circuit resistance: 150©/V
" Length of scale: 110 mm
" Dimensions 154 x 210 x 86mm.
" Weight: 1.5kg
" CEI1010 CATIII 1000Vrms pol2
PSY12
CONTINUOUS - SINGLE PHASE
I (I max = 1,5 x I): 1 - 2A
V (V max = 1,5 x V): 10 - 20 - 50 - 100 - 200 - 400V
3-PHASE
V (V max = 1,5 x V): 380 - 220 - 100 - 50 - 20V
INTENSITY FUSE: 3,15A temp.
Prijs op aanvraag
Meer info ›
Toevoegen aan verlanglijst ›
RMS AC+DC Wattmeter 500mA - 1A (continous, single phase, 3-phase)
PSY11/ELE
The PSY wattmeters measure true RMS and instantaneous power,
DC, single phase, balanced three-phase, alternating signal with
or without DC offset in the frequency range DC up to 2kHz.
ACCURACY : 0.5 %
PROTECTION
" All PSY wattmeters are guaranted within their accuracy limits
for 5 years.
" A fuse to protect the voltage circuit and another for the current.
" The fuses can be accessed from the outside of the meter.
EASY TO READ
" A single scale graduated from 0 to 100.
" The reading on the scale shows the maximum values:
I max = 1.5 x I V max = 1.5 x V
" A lookup table for three phase power can be found on the
back of the unit.
EXAMPLE READING
" For instance we wish to measure the power dissapated in a
load with 0.6A at 125V.
" Put the selector knobs to 1A and 100V.
" The needle moves to the marking corresponding to 75W.
" The reading is direct.
" The 100V scale is not overloaded.
(V max = 1.5 x V therefore 150V).
GENERAL SPECIFICATIONS
" Voltage circuit resistance: 150©/V
" Length of scale: 110 mm
" Dimensions 154 x 210 x 86mm.
" Weight: 1.5kg
" CEI1010 CATIII 1000Vrms pol2
PSY11
CONTINUOUS - SINGLE PHASE
I (I max = 1,5 x I): 500mA - 1A
V (V max = 1,5 x V): 10 - 20 - 50 - 100 - 200 - 400V
3-PHASE
V (V max = 1,5 x V): 380 - 220 - 100 - 50 - 20V
INTENSITY FUSE:2A temp.
Prijs op aanvraag
Meer info ›
Toevoegen aan verlanglijst ›
RMS AC+DC Wattmeter 100-200 mA (continous, single phase, 3-phase)
PSY10/ELE
The PSY wattmeters measure true RMS and instantaneous power,
DC, single phase, balanced three-phase, alternating signal with
or without DC offset in the frequency range DC up to 2kHz.
ACCURACY : 0.5 %
PROTECTION
" All PSY wattmeters are guaranted within their accuracy limits
for 5 years.
" A fuse to protect the voltage circuit and another for the current.
" The fuses can be accessed from the outside of the meter.
EASY TO READ
" A single scale graduated from 0 to 100.
" The reading on the scale shows the maximum values:
I max = 1.5 x I V max = 1.5 x V
" A lookup table for three phase power can be found on the
back of the unit.
EXAMPLE READING
" For instance we wish to measure the power dissapated in a
load with 0.6A at 125V.
" Put the selector knobs to 1A and 100V.
" The needle moves to the marking corresponding to 75W.
" The reading is direct.
" The 100V scale is not overloaded.
(V max = 1.5 x V therefore 150V).
GENERAL SPECIFICATIONS
" Voltage circuit resistance: 150©/V
" Length of scale: 110 mm
" Dimensions 154 x 210 x 86mm.
" Weight: 1.5kg
" CEI1010 CATIII 1000Vrms pol2
PSY10
CONTINUOUS - SINGLE PHASE
I (I max = 1,5 x I): 100 - 200mA
V (V max = 1,5 x V): 10 - 20 - 50 - 100 - 200 - 400V
3-PHASE
V (V max = 1,5 x V): 380 - 220 - 100 - 50 - 20V
INTENSITY FUSE: 315mA temp.
Prijs op aanvraag
Meer info ›
Toevoegen aan verlanglijst ›
RMS AC+DC WATTMETER 10A - 20A (CONTINUE - ENKELFASIG AC)
P7405/ELE
The PSY wattmeters measure true RMS and instantaneous power,
DC, single phase, balanced three-phase, alternating signal with
or without DC offset in the frequency range DC up to 2kHz.
ACCURACY : 0.5 %
PROTECTION
" All PSY wattmeters are guaranted within their accuracy limits
for 5 years.
" A fuse to protect the voltage circuit and another for the current.
" The fuses can be accessed from the outside of the meter.
EASY TO READ
" A single scale graduated from 0 to 100.
" The reading on the scale shows the maximum values:
I max = 1.5 x I V max = 1.5 x V
" A lookup table for three phase power can be found on the
back of the unit.
EXAMPLE READING
" For instance we wish to measure the power dissapated in a
load with 0.6A at 125V.
" Put the selector knobs to 1A and 100V.
" The needle moves to the marking corresponding to 75W.
" The reading is direct.
" The 100V scale is not overloaded.
(V max = 1.5 x V therefore 150V).
GENERAL SPECIFICATIONS
" Voltage circuit resistance: 150©/V
" Length of scale: 110 mm
" Dimensions 154 x 210 x 86mm.
" Weight: 1.5kg
" CEI1010 CATIII 1000V ms pol2
CONTINUOUS - SINGLE PHASE
I (I max = 1,5 x I): 10 - 20A
V (V max = 1,5 x V): 100 - 200 - 400V
INTENSITY FUSE: /
OPTION BOX (OD41) for 3-phase: 400 - 200 - 100V
Prijs op aanvraag
Meer info ›
Toevoegen aan verlanglijst ›
RMS AC+DC Wattmeter 5A - 10A (continous - single phase AC)
P7404/ELE
The PSY wattmeters measure true RMS and instantaneous power,
DC, single phase, balanced three-phase, alternating signal with
or without DC offset in the frequency range DC up to 2kHz.
ACCURACY : 0.5 %
PROTECTION
" All PSY wattmeters are guaranted within their accuracy limits
for 5 years.
" A fuse to protect the voltage circuit and another for the current.
" The fuses can be accessed from the outside of the meter.
EASY TO READ
" A single scale graduated from 0 to 100.
" The reading on the scale shows the maximum values:
I max = 1.5 x I V max = 1.5 x V
" A lookup table for three phase power can be found on the
back of the unit.
EXAMPLE READING
" For instance we wish to measure the power dissapated in a
load with 0.6A at 125V.
" Put the selector knobs to 1A and 100V.
" The needle moves to the marking corresponding to 75W.
" The reading is direct.
" The 100V scale is not overloaded.
(V max = 1.5 x V therefore 150V).
GENERAL SPECIFICATIONS
" Voltage circuit resistance: 150©/V
" Length of scale: 110 mm
" Dimensions 154 x 210 x 86mm.
" Weight: 1.5kg
" CEI1010 CATIII 1000V ms pol2
CONTINUOUS - SINGLE PHASE
I (I max = 1,5 x I): 5 - 10A
V (V max = 1,5 x V): 100 - 200 - 400V
INTENSITY FUSE: 16A temp
OPTION BOX (OD41) for 3-phase: 400 - 200 - 100V
Prijs op aanvraag
Meer info ›
Toevoegen aan verlanglijst ›
RMS AC+DC Wattmeter 2.5 A - 5 A (continous - single phase AC)
P7403/ELE
The PSY wattmeters measure true RMS and instantaneous power,
DC, single phase, balanced three-phase, alternating signal with
or without DC offset in the frequency range DC up to 2kHz.
ACCURACY : 0.5 %
PROTECTION
" All PSY wattmeters are guaranted within their accuracy limits
for 5 years.
" A fuse to protect the voltage circuit and another for the current.
" The fuses can be accessed from the outside of the meter.
EASY TO READ
" A single scale graduated from 0 to 100.
" The reading on the scale shows the maximum values:
I max = 1.5 x I V max = 1.5 x V
" A lookup table for three phase power can be found on the
back of the unit.
EXAMPLE READING
" For instance we wish to measure the power dissapated in a
load with 0.6A at 125V.
" Put the selector knobs to 1A and 100V.
" The needle moves to the marking corresponding to 75W.
" The reading is direct.
" The 100V scale is not overloaded.
(V max = 1.5 x V therefore 150V).
GENERAL SPECIFICATIONS
" Voltage circuit resistance: 150©/V
" Length of scale: 110 mm
" Dimensions 154 x 210 x 86mm.
" Weight: 1.5kg
" CEI1010 CATIII 1000V ms pol2
CONTINUOUS - SINGLE PHASE
I (I max = 1,5 x I): 2,5 - 5A
V (V max = 1,5 x V): 100 - 200 - 400V
INTENSITY FUSE: 8A temp.
OPTION BOX (OD41) for 3-phase: 400 - 200 - 100V
Prijs op aanvraag
Meer info ›
Toevoegen aan verlanglijst ›
RMS AC+DC Wattmeter 1A - 2A (continous - single phase AC)
P7402/ELE
The PSY wattmeters measure true RMS and instantaneous power,
DC, single phase, balanced three-phase, alternating signal with
or without DC offset in the frequency range DC up to 2kHz.
ACCURACY : 0.5 %
PROTECTION
" All PSY wattmeters are guaranted within their accuracy limits
for 5 years.
" A fuse to protect the voltage circuit and another for the current.
" The fuses can be accessed from the outside of the meter.
EASY TO READ
" A single scale graduated from 0 to 100.
" The reading on the scale shows the maximum values:
I max = 1.5 x I V max = 1.5 x V
" A lookup table for three phase power can be found on the
back of the unit.
EXAMPLE READING
" For instance we wish to measure the power dissapated in a
load with 0.6A at 125V.
" Put the selector knobs to 1A and 100V.
" The needle moves to the marking corresponding to 75W.
" The reading is direct.
" The 100V scale is not overloaded.
(V max = 1.5 x V therefore 150V).
GENERAL SPECIFICATIONS
" Voltage circuit resistance: 150©/V
" Length of scale: 110 mm
" Dimensions 154 x 210 x 86mm.
" Weight: 1.5kg
" CEI1010 CATIII 1000V ms pol2
CONTINUOUS - SINGLE PHASE
I (I max = 1,5 x I): 1 - 2A
V (V max = 1,5 x V): 100 - 200 - 400V
INTENSITY FUSE: 3,15A temp.
OPTION BOX (OD41) for 3-phase: 400 - 200 - 100V
Prijs op aanvraag
Meer info ›
Toevoegen aan verlanglijst ›
RMS AC+DC Wattmeter 500mA-1A (continous - single phase AC)
P7401/ELE
The PSY wattmeters measure true RMS and instantaneous power,
DC, single phase, balanced three-phase, alternating signal with
or without DC offset in the frequency range DC up to 2kHz.
ACCURACY : 0.5 %
PROTECTION
" All PSY wattmeters are guaranted within their accuracy limits
for 5 years.
" A fuse to protect the voltage circuit and another for the current.
" The fuses can be accessed from the outside of the meter.
EASY TO READ
" A single scale graduated from 0 to 100.
" The reading on the scale shows the maximum values:
I max = 1.5 x I V max = 1.5 x V
" A lookup table for three phase power can be found on the
back of the unit.
EXAMPLE READING
" For instance we wish to measure the power dissapated in a
load with 0.6A at 125V.
" Put the selector knobs to 1A and 100V.
" The needle moves to the marking corresponding to 75W.
" The reading is direct.
" The 100V scale is not overloaded.
(V max = 1.5 x V therefore 150V).
GENERAL SPECIFICATIONS
" Voltage circuit resistance: 150©/V
" Length of scale: 110 mm
" Dimensions 154 x 210 x 86mm.
" Weight: 1.5kg
" CEI1010 CATIII 1000V ms pol2
CONTINUOUS - SINGLE PHASE
I (I max = 1,5 x I): 500mA - 1A
V (V max = 1,5 x V): 100 - 200 - 400V
INTENSITY FUSE: 2A temp.
OPTION BOX (OD41) for 3-phase: 400 - 200 - 100V
Prijs op aanvraag
Meer info ›
Toevoegen aan verlanglijst ›
ZEKERING œF 30 (250V - 50MA) VOOR PSY WATTMETERS
UF30/ELE
Prijs op aanvraag
Meer info ›
Toevoegen aan verlanglijst ›
MOVING-COIL VOLTMETER (TRUE RMS) 75-150-300-600V
P7476/ELE
Prijs op aanvraag
Meer info ›
Toevoegen aan verlanglijst ›
WATTMETER DIGITAAL (Volt, Amp, Watt)
WATTELEC/ELE
Prijs op aanvraag
Meer info ›
Toevoegen aan verlanglijst ›
WATTMETER SINGLE-PHASE - DC - 3-PHASE - W 17
W17/ELE
Single-phase / DC / 3-phase watt meter, single current rating at 5 amps. Only 2 easy reading levels with a parallax mirror. It is compatible with the watt meter switch using a 2 watt meter measurement method with separate voltage and intensity circuits.
Selector calibers
Monophasic and continuous voltage: 60 120 180 240 360 480
Volts3-phase voltage: 100 200 310 415 Volts
" Accuracy: 2.5% in DC and 1% in AC
" AC frequency range: 4565Hz
" Protection with an HPC fuse on the current circuit
" Auto-protected on the voltage circuit
" Double insulation
" CEI-1010 CAT III 600V (pol2)
Prijs op aanvraag
Meer info ›
Toevoegen aan verlanglijst ›